9. A correct installation will usually show positive kW for all 3 phases and positive kVAr for all three phases.
If negative values are shown it is probable that one or more of the coils are either the wrong way around or on the wrong phases. Colour coded cables are sometimes incorrectly marked.
After correcting the coil installation, repeat the autophasing procedure from step 3.
Once successfully installed, the Profile system will record data relating to all three connected phases as well as the voltage reference.
When the survey is complete, data may be downloaded to a PC using the ProPower 3 software as detailed in the software documentation.
If required the Profile can also be used as a simple meter by reading the kWh & kVArh registers at the start and end of a survey to provide total circuit consumption for the survey period.
This document is intended as a guide only. NewFound Energy cannot be held responsible for incorrect data recordings caused by incorrect Profile installation.
Any customer requiring further guidance in the use of Profile should contact their supplier.